ISO. Second edition. Reference number Permission can be requested from either ISO at the address below or ISO’s member body in the. 31 Jul Purchase your copy of BS EN ISO as a PDF download or hard copy directly from the official BSI Shop. All BSI British Standards. EN ISO Intravascular catheters – Sterile and single-use catheters – Part 5 : Over-needle peripheral catheters – ISO specifies requirements.
|Published (Last):||2 August 2007|
|PDF File Size:||19.73 Mb|
|ePub File Size:||7.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
An approved test method for producing a value of radio-detectability will be established.
Fitting attached to the needle tube, providing communication with its bore. An example of a suitable apparatus is shown in iso 10555-5 ure B.
The catheter tube may have a single lumen or isoo lumens. No part of the these publications may be reproduced in any iso 10555-5 without the prior permission in writing of BIS.
Iso 10555-5 corresponding Indian Standards izo are to iso 10555-5 substituted in their places are listed below along with their degree of equivalence for the editions indicated: International Standard ISO Other parts of this standard are as follows: Blood flow into the needle hub. Assembly comprising at least a needle tube attached to, and communicating with, a needle hub. Review of Indian Standards Amendments are issued to standards as the need arises on the basis of comments.
BS EN ISO 10555-5:2013
The fluid is allowed to flow into the needle, and the time taken for fluid to leak through the vent fitting is measured. Page 3, subclause 4. E-5 Test report The test report shall iso 10555-5 at least the following iso 10555-5 The number of significant places retained in the rounded off value should be the same as that of the specified value in this standard.
Search the history iso 10555-5 over billion web pages on the Internet. Limited collection of colours.
All standards are subject to revision, and parties to agreements based on this part of ISO are encouraged to investigate the possibility of applying the most iso 10555-5 editions of the standards indicated below. Figure 1 — Typical over-needle peripheral intravascular catheter NOTE 2 At the time of publication of this part of ISOthere is no acceptable, validated iso 10555-5 method to determine iso 10555-5.
Details of which is given in National Annex A. Add the following reference: This does not preclude the free use, in the course of implementing the standard, of necessary details, such as symbols and sizes, type or iso 10555-5 designations.
Intravascular catheters ISO , -3, -4, -5
100555-5 the time taken for fluid to form the first falling drop at the back of the vent fitting. If a steel tube is used, it shall connply with ISO 1055-5 The fluid pathway in the needle shall be free of unintended obstructions that would prevent flashback. Round the calculated average water fiowrate to the nearest whole number iso 10555-5 millilitres.
Catheter designed for the introduction or withdrawal iso 10555-5 liquids or devices into or from the peripheral vascular system.
I Principle Water is allowed to flow through the catheter and the amount of flow is measured either volumetrically or gravimetrically. At the time of publication, the editions 1055-5 were valid. Part 1 General requirements do The technical committee has reviewed ido provisions of 1055-5 following International Standard referred iso 10555-5 this adopted standard and has decided that it is acceptable for use in conjunction with this standard: For the purpose of deciding whether a particular requirement of this standard is complied with, the final value, observed or calculated expressing the result of a test or analysis, shall be rounded off in accordance with IS 2: Annex D shows examples of typical needle point geom- etries.
Rigid iso 10555-5 with one end sharpened to facilitate entry into body tissue. Certain conventions are, however, not identical to those used in Indian Standards. The assembly comprising at least a needle tube attached iso 10555-5 and communicating with a needle hub.